Zeiss LSM 510 Meta inverted confocal manual - Research Programs. A heating chamber covering the whole microscope and a CO2-controller makes long term timelapses feasible, cells have survived for more than a weekend under these conditions. Zeiss LSM 510 Meta inverted confocal manual. *adapted from. have been left on for you by the previous user, as long as anyone scheduled after you won't use.
LSM 510 / LSM 510 META Upgrade ZEN 2009 The movie to the rht shows cells happily dividing and moving around during a time period of 16 hours. Specifications Equipped with 4 lasers, 2 regular detectors (Ps) and a META detector, which may also serve as an additional detector. Information available in the full user manual or in the manual of the. Double click the ZEN 2009 icon on the WINDOWS desktop to start the Carl Zeiss LSM.
LSM 510 / LSM 510 META Upgrade ZEN 2009 - MCD Biology The low magnification/NA objectives are of hh quality making the miroscope well suited for also lager specimens, e.g. A transmission detector enables confocal "Pseudo-DIC". A heating chamber and a CO2 controller fasilitates live time experiments. Carl Zeiss. 10/2009. M60-1-0019 e. III. How to make best use of the LSM 510 and LSM 510 META operating instructions This operating manual also includes.
Confocal LSM510 User Guide Pdf - University of Bath In addition a CCD-camera is mounted for regular widefield fluorescence and brhtfield imaging. Manual and certainly does not cover all ques, such as line scans and DIC imaging. the Zeiss manual in the confocal room or from experienced confocal users. Double click on the LSM 510 icon to start confocal operating software.
LSM 510 and LSM 510 META Laser Scanning Microscopes Lasers / Laser lines: Blue diode (BD) - 25 m W / 405Argon (Ar) - 30 m W / 458, 472, 488, 514 Helium/Neon I (He Ne I) - 1 m W / 543Helium/Neon II (He Ne II) - 5 m W / 633 Objectives: M: magnification NA: numerical aperture IM: immersion medium WD: working distance Links LSM 5 Image Browser (Fill in the form and submit. LSM 510. Carl Zeiss. LSM 510 META. II. B 45-0008 e 10/02. Knowledge of this manual is required for the operation of the instrument. Would you therefore.
Zeiss simplified LMS-510 guide Room 6C112c ATelephone 555 86581 This confocal system is reliable working horse and well equipped for a broad range of dyes, long term timelapse and emission fingerprinting (META detector). The LSM 510 and LSM 510 META laser scanning microscope, including its. In the Operating Manual, read the chapter Safety Instructions carefully before.
Zeiss LSM 510 Confocal Microscope Operating Manual - Research The most common samples are fixed cells and tissue and 4 seperate colors can easily be combined. ZEISS LSM 510. CONFOCAL MICROSCOPE. OPERATING INSTRUCTIONS. Last Revision 5/21/2009. 1. ZEISS LSM 510 POWER ON/OFF PROCEDURES.
LSM 510 Laser Scanning Microscope How to make best use of the LSM 510 operating instructions. Please contact Carl Zeiss if you intend to use a laser type with a wavelength other than the.
Guided Tour For LSM 510 And Meta Inverted Zeiss LSM 510 META confocal microscope. Axiovert 200M. Starting the Zeiss LSM 510 microscope, software and laser. for detailed instructions!
Manual LSM 510 Meta 0- LNMA Multiphoton Laser Scanning Microscopy - Using the Zeiss LSM 510 META NLO. 10 Certification. 11 Brief Operating Manual. This section.